Can data be saved at once with the QIAxcel System after analysis of all samples in a 96-well PCR plate?
The QIAxcel System is set up to analyze samples in rows (A to H) and saves the data after separation for each row (12-channel cartridge setting). To automatically analyze all samples in a 96-well PCR plate:
- In the Instruments Control Panel dialog box, select A from the "Pos" drop-down list.
- Enter 8 in the "Runs" field.
- Select the "Inc." (Increment) check box.