Cat. No. / ID: 9002864
QIAcube Connect lets you automate your manual spin-column-based sample preparation kits right out of the box and purify up to 12 samples in a single run.
To work even faster, get kits specifically designed for use with QIAcube Connect. These come with spin columns and tubes already loaded onto the rotor, so you don't have to put them in yourself:
QIAcube Connect operates in conjunction with QIAcube Connect Accessories.
For QIAcube resources, visit the QIAcube (classic) resource page.
The plug-and-play QIAcube Connect simplifies purification by automating QIAGEN spin-column kits. No change of chemistry is needed, so startup is fast, results are immediate, and performance is comparable to the manual procedure.
QIAcube Connect includes a centrifuge, heated shaker, pipetting system and robotic gripper (see " The QIAcube Connect worktable"). These let the QIAcube Connect fully automate more than 80 QIAGEN spin column kits.
UV light decontamination
QIAcube Connect decontaminates your full worktable with built-in UV light (see " The QIAcube Connect UV light") to prevent carryover between samples (see " Genomic DNA (gDNA) barely detectable after UV light exposure").
Remote connectivity with QIAsphere
QIAcube Connect is QIAsphere-ready. QIAsphere gives full connectivity so you can step out of the lab to ponder on your research while keeping an eye on your instrument run status.
Stay alert to important instrument events with QIAsphere's push notifications. You can even check maintenance status and manage run reports from anywhere, at anytime. Increase the efficiency and productivity of your lab. Learn more about QIAsphere.
Track samples and sample prep kit information throughout the process, using the handheld barcode scanner that comes with the kit.
Light inside the instrument makes it easier for you to load your samples and perform your experiments.
Protocols for various sample types
QIAcube Connect has a variety of pre-installed protocols for purification of RNA, genomic DNA, plasmid DNA, viral nucleic acids and proteins, plus DNA and RNA cleanup. All standard protocols in the expanding range can also be downloaded free of charge. Customized protocols can also be requested.
QIAcube Connect isolates highly pure nucleic acids or proteins using a spin-column-based lyse, bind, wash and elute procedure (see '' Automated QIAprep Mini Standard procedure'').
A large touchscreen (see figure '' The QIAcube Connect touchscreen'') simplifies protocol selection. Clear on-screen messages guide you through worktable setup. You are unlikely to make mistakes because the labware and accessories fit onto the worktable only in the correct orientation. In addition, a fully automated load check helps to ensure that samples, reagents and labware are loaded correctly.
Dedicated QIAcube Connect kits further simplify automated spin-preps for purification of RNA, genomic DNA and viral RNA. Rotor-adapters that come with the kits are pre-loaded with spin columns and elution tubes, so they are ready to use right out of the box. And then, to lessen the risk of errors even more, tubes required for the manual procedure are not included in kits dedicated for use with QIAcube Connect.
QIAcube Connect is highly suitable for academic research laboratories as well as pharmaceutical, biotechnology, and biomedical research laboratories performing applications, such as:
Have your throughput demands increased? Do you want to expand your application range or further streamline workflows? Do you need to manage more complex samples? Do you need improved connectivity?
If the answer to any of these questions is yes, then QIAGEN’s trade-in/trade-up program is just right for you!
QIAGEN makes it easy to keep up to speed with the latest automation and sample processing technologies. Simply exchange an old instrument for a new one. Improve your efficiency by trading in a competitor instrument or trading up with an older QIAGEN instrument.
|Technical data||Power: 100–240 V AC, 50/60 Hz, 650 VA Mains supply voltage fluctuations are not to exceed 10% of nominal supply voltages. Note: The apparent power can exceed 650 VA for up to 2 seconds during the centrifuge acceleration and can reach an approximate value of 1000 VA.; Fuse: 2x T6.3A L 250V; Overvoltage category: II|
|Operating temperature||18 to 28ºC (64.4 to 82.4ºF)|
|Relative humidity||15–75% (non-condensing)|
|Place of operation||For indoor use only|
|Environmental class||3K2 (IEC 60721-3-3)|
|Storage conditions||Temperature: 15ºC to 30ºC (59ºF to 86ºF) in manufacturer’s package; Relative humidity: Max. 75% (non-condensing); Environmental class: 1K2 (IEC 60721-3-1)|
|Instrument dimensions||Dimensions (hood closed): width 65 cm (25.6 in.), height 58 cm (22.8 in.), depth 62 cm (24.4 in.); Dimensions (hood open): width 65 cm (25.6 in.), height 86 cm (34 in.), depth 62 cm (24.4 in.); Mass: QIAcube Connect: 71.5 kg (157.6 lb.), Accessories: 3 kg (6.6 lb.)|
|Centrifuge||10,640 rpm maximum, 12,000 x g maximum, swing-out rotor maximum 45°, 12 rotor positions|
|Shaker||Speed 100–2000 rpm, amplitude 2 mm, heating range of ambient temperature to 70°C (158°F), ramp-up time of <5 minutes from ambient temperature to 55°C (±3°C). Difference in the temperature detected by the internal sensor and the temperature of the sample liquid is approximately –2°C.|
|Pipetting system||Syringe size 1 ml, pipetting range 5–900 µl|
|Table capacity||Up to 12 samples per run|
|Touchscreen||10.4” TFT Touchscreen, active area 211.2 x 158.4 mm, resolution 800*600 SVGA|
|Data output||USB stick: USB2.0; Compatible OS: Windows 8, Windows 7, Windows Vista, Windows XP (SP3 or later); Mac OS x 10,1 or later; Operating temperature range: 0ºC to 35ºC; Operating humidity range: 10 to 90% (with no condensation); Storage temperature range: –20ºC to 60ºC (–4ºF to 140ºF); Storage humidity range: 10 to 90% (with no condensation); Formatting: FAT32|
|UV lamp||Wavelength: 278nm; Optical power: 200-300mW|
|Special feature||Handheld scanner; Scan pattern: Area Image (838 x 640-pixel array); Motion tolerance: Up to 610 cm/s (240 in/s) for 13 mil UPC at optimal focus; Symbol contrast: 20% minimum reflectance difference; Decode capability: Reads standard 1D, PDF, 2D, Postal and OCR symbology|
|Software||QIAGEN protocols are pre-installed on the QIAcube Connect.|
|Altitude||Up to 2000 m (6500 ft)|
|Transportation conditions||–25°C to 60°C (–13°F to 140°F) in manufacturer's package; Max. 75% relative humidity (non-condensing); Environmenal class: 2K2 (IEC 60721-3-2)|
Cat. No. / ID: 9245209
Cat. No. / ID: 9245260
Cat. No. / ID: 9245210
Cat. No. / ID: 9245208
Yes. The heated shaker on the QIAcube can be operated independently if the QIAcube is not performing a protocol run. For instructions on how to operate the shaker, refer to the QIAcube User Manual, chapter 5 'Operating Procedures, section 5.7 'Operating the shaker'.
The shaker can be used to heat and shake samples simultaneously, shake samples without heating, or heat samples without shaking.
See: http://www.qiagen.com/products/catalogpagecontrols/header/qiacube/360/default.aspx for a virtual demonstration of the heater and shaker on the QIAcube.
Visit the QIAcube Protocols page to search for protocols. The instructions for transferring a protocol to the QIAcube are listed below.
To transfer new protocols to the QIAcube, first save them to the supplied USB stick.
Note: Protocols already installed on the QIAcube that have the same name as downloaded protocols will be overwritten.
8. When the protocols have been successfully transferred to the QIAcube, press “OK”.
Yes. At the end of each protocol run on the QIAcube, a report file is generated that contains important information such as number of samples, liquid volumes measured during the load check, and run time. A log file is also generated that contains information about instrument actions, such as movement of the robotic arm.
Yes. The QIAcube centrifuge can be operated independently if the QIAcube is not performing a protocol run. For instructions on how to operate the centrifuge, refer to the QIAcube User Manual, chapter 5 'Operating Procedures, section 5.8 'Operating the Centrifuge'.
Yes, this is absolutely necessary. If the lids are not securely fixed in the rotor adapter slots during processing on the QIAcube, they will become detached from the elution tubes and columns during centrifugation and will collect at the bottom of the centrifuge chamber.
The following disinfectants and detergents are recommended for cleaning the QIAcube:
Lysetol® AF (Gigsept® Instru AF) (Schülke & Mayr GmbH) — disinfectant for submerging worktable items (consists of 14 g cocospropylene-diamine-guanidine diacetate, 35 g phenoxypropanols, and 2.5 g benzalkonium chloride per 100 g Lysetol AF, with anticorrosion components, fragrance, and 15–30% nonionic surfactants).
DECON-QUAT® 100 (Veltek Associates, Inc.) — quaternary ammonium salt based disinfectant concentrate for submerging worktable items (contains 5% alkyldimethylbenzylammonium chloride and 5% alkyldimethylethylbenzylammonium chloride).
Note: If you want to use disinfectants different from those recommended, ensure that their compositions are similar to those described above.
No. The QIAcube is designed for fully automated processing of up to 12 samples using QIAGEN spin-column kits. Do not use spin columns manufactured by other suppliers with the QIAcube. Damage caused by use of other types of spin columns will void the instrument warranty.
See: http://www.qiagen.com/products/catalogpagecontrols/header/qiacube/360/default.aspx for a demonstration on how the QIAcube is used with the RNeasy Mini Kit for automated purification of RNA.
Protocols for automated DNA extraction from saliva stabilized with the PAXgene Saliva Collector are available for the QIAsymphony DNA Midi Kit with the QIAsymphony SP instrument or the QIAamp DNA Mini Kit with the QIAcube (Classic and Connect).
For manual extraction, supplementary protocols are available for the QIAamp DNA Mini and Gentra Puregene Cell Kits (see resources section).
No. Modified QIAGEN protocols or custom protocols developed to meet your specific requirements can be requested and purchased from QIAGEN.
1. Go to www.qiagen.com/MyQIAcube and enter the required information. If our Application Specialists require additional information, they will contact you directly.
2. You will be informed by e-mail when your customized protocol is ready for download.
3. Click the link in the e-mail to go to the download page.
See: http://www.qiagen.com/Products/Catalog/Automated-Solutions/Sample-Prep/QIAcube#orderinginformation for a virtual demonstration on how to use the QIAcube.
The QIAcube has the following weight and dimensions:
Please refer to section 6.5 'Decontaminating the QIAcube' in chapter 6 'Maintenance Procedures' of the QIAcube User Manual.
You can stop a QIAcube protocol if there is an emergency by pressing “Cancel”. To confirm that you want to stop the protocol run, press “OK”. To cancel the 'stop protocol' command, press “Cancel”.
Note: If a protocol run is stopped, the run cannot be restarted. The samples must be processed manually. The last protocol step performed by the QIAcube will be displayed in the touch screen. For more information, refer to the QIAcube User Manual.
The corresponding manual kits can be used for all available protocols on the QIAcube. The manual kits contain sufficient buffers and reagents to process the number of spin columns provided with the kit.
In addition to the purification kit, Rotor Adapters (10 x 24), cat. no 990394, and the required disposable tips:
need to be ordered separately. To find out which filter-tips are required for a specific application, refer to the corresponding protocol sheet.
No. The QIAcube is equipped with an optical sensor, which determines the tip size and the number of tips loaded onto the worktable prior to starting a run.
The QIAcube can use partially used tip racks from previous runs, provided that the racks contain sufficient tips to process the number of samples loaded for the current run.