Cat. No. / ID: 950067
The QIAamp 96 PowerFecal QIAcube HT Kit provides simple, automated purification of high-quality genomic DNA (human and bacterial) from up to 200 mg fresh or frozen stool samples using the QIAcube HT. The novel combination of MOBIO's patented Inhibitor Removal Technology (IRT) and proven QIAamp purification system provides an optimized DNA purification solution for microbiome, infectious disease and metabolic research applications.
The QIAamp 96 PowerFecal QIAcube HT Kit is to be discontinued shortly. Try the DNeasy 96 PowerSoil Pro QIAcube HT Kit – for improved performance. Higher yields, better purity, more accurate representation of the microbial diversity.
|Time per run||96 samples in approximately 120 minutes 24 samples in approximately 60 minutes|
|Sample amount||Up to 200 mg fresh or frozen stool|
|Throughput||24-96 samples in increments of 8 samples|
|Elution volume||100 μl|
No, the two software applications are not interchangeable.
Additionally, the QIAcube HT and QIAxtractor software cannot be installed on the same computer. The QIAcube HT software versions is v4.17 or higher, while the QIAxtractor software version is v4.16 or lower.
No, this upgrade is not possible.
The Support Packages are important when reporting an issue to QIAGEN Technical Services. Upon completion of each run, the software automatically saves a support package.
To retrieve a support package, click Help in the QIAcube HT or QIAxtractor software and select Choose support package. In the new windows, select the support package that corresponds to the issue and send it to QIAGEN Technical Services.
1. Check the vacuum tubing to make sure it is not pinched or obstructed in some way. Sometimes the tubing can become pinched against the bench top when the vacuum station is placed too far from the instrument.
2. Make sure that the channeling adapter is properly placed. Check if the carriage is moving the plate completely from the waste chamber position to the elution chamber position. If the calibration seems incorrect, the instrument may need to be re-taught at this position.
3. When using less than a full plate, make sure all the unused columns are covered with an adhesive seal.
Also, make sure each column of 8 samples is full. When processing a number of samples that results in < 8 samples per column, use water as a substitute in the empty positions until the column is filled. This ensures even vacuum across all unsealed well positions.
4. Make sure correct plates are used for the selected protocol.
5. When the vacuum is on, please listen for any hissing noise from the tubing that is coming down to the waste container. This could indicate leaks in the tubing.
If the issue persists, please contact QIAGEN Technical Service.
Make sure the QIAcube HT or QIAxtractor software uses the same COM port number as assigned to the QIAcube HT or QIAxtractor instrument on the computer. The COM port setting can be checked by clicking Options\Robot Setup\Select COM port in the QIAcube HT or QIAxtractor software.
Ensure that the cable between the instrument and the computer is securely inserted. Power on the instrument first, and then launch the QIAxtractor/QIAcube HT software.
The instrument has both RS-232 (serial) and USB interfaces for computer connection. Try an RS-232 cable if the communication cannot be established between the instrument and computer with a USB cable.
If the issue persists, please contact QIAGEN Technical Services.